Message 7 of If you are sure that you have selected the correct instrument and it is responding, try disabling the ID Query. Message 12 of Mi whole acquisition programm seems to be working Yes I had it on the Network when I power cycled the E And I have also read that thread u mention before I asked this question.
|Date Added:||21 February 2011|
|File Size:||44.3 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
As a side note on some of the older Agilent s4438c I believe it is the LINUX based operating systems modelsyou have to make sure that the ethernet port is plugged into an active network or the ethernet drivers don’t get loaded when you cycle power. I don’t know why you couldn’t upgrade the firmware. I haven’t implemented this because it’s a bit of a change to some legacy code and I haven’t had time to experiment with the implementation.
I don’t remember the exact process but you should select the manual ailent for creating the resource and enter the ip address. Thank you for answering. Just remember to close it later. Repeat immediately from the beginning or do something else? I e443c8 your help.
If it doesn’t find the instrument ID, I would be suspicious of your connection. I look through the VI in the instrument driver but I can’t seem to find it.
Message 5 of 8. Thanks for the suggestion. Message 18 of Your code example gave me a clue to where I was going wrong.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. You need to know the VISA resource name of the Agilent device albview order to start communicating with the device. My support system ensures that I don’t look totally incompetent.
Agilent ESG Series Signal Generator
The drivers are bugged in my opinion. Message 12 of Looks to me like it is searching for Agilent or Keysight ID codes. Sorry I am new to this commands how do I use agiilent commands? Message 19 of I can type it in labview?
EC auto frequency change – Discussion Forums – National Instruments
Message 4 of I needed to use the as the port. Message 4 of 8.
Here’s the search results from there: Do you have a manual or data sheet for the device that has these commands? Yes I had it on the Network when I power cycled the E Thank you in advance. But, yes, I think so.
I’ve never had much luck with RAW resources but it’s worth a try. I’ll give Keysight a call and see if they know. Most Active Software Boards: Proud to say that I’ve progressed beyond knowing just enough to be dangerous.
This way it doesn’t require the.